CAUTION: Follow disi nfectant
manufacturer's rinsing instr uctions.
Fill basin with disinfectant. Attach channel plug
and injection tube to previously cleaned
endoscope. Immerse completely in disinfectant
While the endoscope is immersed,
disconnect cleaning accessories and use
a clean lint-free cloth to wipe away any
bubbles clinging to external surfaces.
Use syringe to completely fill air, water and
suction channels with disinfectant solution.
Soak all equipment according to the
disinfectant manufacturer's recommended
time and temperature. Use a timer to
measure exact contact time.
Connect channel plug and injection tube.
Inject air to remove disinfectant from air,
water and suction channels. Remove
endoscope from disinfectant.
Disconnect channel plug and injection
tube. Immerse endoscope and
detached parts in water. Thoroughly
rinse all external surfaces.
Connect channel plug and injection tube to
endoscope. Use a syringe to flush water
through air, water and suction channels.
Remove endoscope from rinse water. Use
syringe to inject air through air, water and
suction channels, expelling the rinse water.
Use syringe to flush 70% alcohol through
air, water & suction channels. Inject air
to remove retained alcohol.
TRANSPORTATION•Use a compatible disinfectant.Test for proper concentrationper manufacturer's instructions. Clean endoscope meticulouslyprior to disinfection.Autoclave all autoclavableaccessories.
•Store endoscope hangingvertically with valves and capsremoved, and with angulationlock in a “free” position.
Hold all parts of the endoscope when
carrying. Do not let the ends of the
endoscope swing freely.
Disconnect all accessories from the endoscope.
Dry endoscope and accessories with an alcohol-
moistened cloth. Make sure equipment is
completely dry prior to storage.
2003 Olympus America, Inc. / Printed in USA - 08/03/ PWC-100002
ENDOSCOPE DISINFECTION GUIDE
™ PEF-V Transnasal Esophagovideoscope
Refer to compa nion ENDOSCOPE CLEANING GUIDE fo r cleaning information