• The lower limit is stored in the channel.
7Select the desired frequency as the upper limit.
8Press [F], [MR].
9Press [UP]/ [DWN] to select a matching channel
in the range U0 to U9.
• If you have selected for example L3 in step 5, select
10 Press [OK].
• The upper limit is stored in the channel.
To confirm the stored scan limits, press [MR], then
select the L and U channels.
The lower limit must be lower in frequency than the upper limit.
The lower and upper frequency step sizes must be equal.
The lower and upper limits must be selected on the same band.PROGRAM SCAN
Program Scan is identical with VFO Scan except that
you select the frequency range of the scan.
■Setting Scan Limits
You can store up to 10 scan ranges in memory
channels L0/U0 to L9/U9.
1Select the desired band.
3Select the desired frequency as the lower limit.
4Press [F], [MR].
5Press [UP]/ [DWN] to select a channel in the
range L0 to L9.
TH-D7 E 33 Scan 98.12.4, 0:11 PM37